Safety valve test and calibration bench-Manual control/computer control model 1. Product application and functional characteristics Safety valve test bench is used for various safety valve setting pressure adjustment, the whole set pressure detection, discharge pressure detection, back seat...
Safety Valve Test Bench-Manual control/computer control model
Here below is the video link for manual+digital control safety valve test bench:
Herebelow is the video link for portable safety valve test bench:
The safety valve test bench is a new type of pressure test device developed by our company for reference to GB/T 12242-2005 (pressure release device and performance test specification), which is used to adjust and test the setting pressure of the safety valve.
2, Function and features
Safety valve test benchis used for various safety valve setting pressure adjustment, the whole set pressure detection, discharge pressure detection, back seat pressure testing, assessment of its pressure safety performance etc.. ,
A. pressure range: 0-480 bar for option(gas pressure and water pressure)
B. test medium: air, nitrogen, water;
C. test valve diameter: DN15-DN400 for option
D. pressure control accuracy: less than 1%F.S;
E. pressure measurement accuracy: 0.4 pressure gauge (optional higher pressure level);
F. Clamping way: hydraulic cylinder clamping;
G. station interface type: screw and flange type
H. clamping force output range: 0-50 tons;
I. supply voltage: 50Hz -60Hz ,220V/380V for option
J:Size：1500×1200×900mm(machine) +1500×800×800mm(clamping device)
5.Instruction for software control part
The first step: start the computer and double click the boot program, as shown in Figure 1.
The second step: 1) after the program, enter the main interface of the program.
2）In the main interface, select the appropriate sensor, and pay attention to the selection of the sensor not to exceed the 2/3. of the measuring process ,as shown in Figure3
3) Display the dialog box for the test information, as shown below
With printer,could print the testing report directly.